Unexpected rhythms with normally functioning dual-chamber pacing systems
- David L Hayes, MD
David L Hayes, MD
- Professor of Medicine
- Mayo Medical School
A variety of electrocardiographic findings and rhythms may be encountered in which the pacing system is functioning normally. These include crosstalk, pacemaker-mediated or endless loop tachycardias, and repetitive non-reentrant ventriculo-atrial synchronous rhythms. These might be considered functional system malfunctions since the resulting arrhythmia may be deleterious to the patient, although the device is functioning normally and in accord with its programmed specifications.
A comprehensive discussion of the normal function and programming of permanent pacemakers is presented separately. (See "Permanent cardiac pacing: Overview of devices and indications" and "Modes of cardiac pacing: Nomenclature and selection".)
In a DDD pacemaker, a paced or sensed event in one channel initiates one or more timing circuits in the opposite channel. As a result, the release of an atrial output pulse will initiate an atrioventricular (AV) delay, which will permit a ventricular output pulse at the end of this interval if an R wave is not sensed. A ventricular event that is sensed during the AV interval, considered to be an R wave, will cancel the AV interval, inhibit the ventricular output pulse, and cause the pacemaker to initiate an atrial escape interval.
Current pacing systems, however, are incapable of recognizing morphologic characteristics of a complex. Any intrinsic complex, output decay, or artifact of electromagnetic interference of sufficient amplitude and frequency response that it is not filtered out by the sense amplifier is treated as a true signal. Although bipolar sensing configuration is believed to minimize far-field sensing, crosstalk can be seen with any sensing configuration. If any signal is interpreted as an R wave, and inhibits the ventricular output and resets the escape interval, the phenomenon is termed crosstalk-mediated ventricular output inhibition.
When AV nodal conduction is otherwise intact, crosstalk is asymptomatic. If there is first degree AV block, the effective paced rate may be slower than the programmed base rate, because the conducted R wave will occur after completion of the ventricular refractory period initiated by the oversensing of the atrial stimulus. The conducted R wave will be sensed, thereby resetting the atrial escape interval yet again. A clinical catastrophe may occur in the presence of complete AV block combined with crosstalk (in the absence of safety-pacing), as the system will effectively pace the atrium, but the ventricular output may be repeatedly inhibited, leaving the patient asystolic.
Subscribers log in hereLiterature review current through: Nov 2017. | This topic last updated: Sep 14, 2017.References
- Stroobandt RX, Barold SS, Vandenbulcke FD, et al. A reappraisal of pacemaker timing cycles pertaining to automatic mode switching. J Interv Card Electrophysiol 2001; 5:417.
- Luceri RM, Ramirez AV, Castellanos A, et al. Ventricular tachycardia produced by a normally functioning AV sequential demand (DVI) pacemaker with "committed" ventricular stimulation. J Am Coll Cardiol 1983; 1:1177.
- Frumin H, Furman S. Endless loop tachycardia started by an atrial premature complex in a patient with a dual chamber pacemaker. J Am Coll Cardiol 1985; 5:707.
- Van Gelder BM, Bracke FA, Meijer A. Pacemaker-mediated tachycardia in a biventricular pacing system. Pacing Clin Electrophysiol 2001; 24:1819.
- Barold SS, Byrd CL. Cross-ventricular endless loop tachycardia during biventricular pacing. Pacing Clin Electrophysiol 2001; 24:1821.
- Klementowicz PT, Furman S. Selective atrial sensing in dual chamber pacemakers eliminates endless loop tachycardia. J Am Coll Cardiol 1986; 7:590.
- Fröhlig G, Schwerdt H, Schieffer H, Bette L. Atrial signal variations and pacemaker malsensing during exercise: a study in the time and frequency domain. J Am Coll Cardiol 1988; 11:806.
- Barold SS, Levine PA. Pacemaker repetitive nonreentrant ventriculoatrial synchronous rhythm. A review. J Interv Card Electrophysiol 2001; 5:45.
- Chien WW, Foster E, Phillips B, et al. Pacemaker syndrome in a patient with DDD pacemaker for long QT syndrome. Pacing Clin Electrophysiol 1991; 14:1209.
- Nielsen JC, Kristensen L, Andersen HR, et al. A randomized comparison of atrial and dual-chamber pacing in 177 consecutive patients with sick sinus syndrome: echocardiographic and clinical outcome. J Am Coll Cardiol 2003; 42:614.
- Sweeney MO, Hellkamp AS, Ellenbogen KA, et al. Adverse effect of ventricular pacing on heart failure and atrial fibrillation among patients with normal baseline QRS duration in a clinical trial of pacemaker therapy for sinus node dysfunction. Circulation 2003; 107:2932.
- Wilkoff BL, Cook JR, Epstein AE, et al. Dual-chamber pacing or ventricular backup pacing in patients with an implantable defibrillator: the Dual Chamber and VVI Implantable Defibrillator (DAVID) Trial. JAMA 2002; 288:3115.
- O'Keefe JH Jr, Abuissa H, Jones PG, et al. Effect of chronic right ventricular apical pacing on left ventricular function. Am J Cardiol 2005; 95:771.
- Shukla HH, Hellkamp AS, James EA, et al. Heart failure hospitalization is more common in pacemaker patients with sinus node dysfunction and a prolonged paced QRS duration. Heart Rhythm 2005; 2:245.
- Melzer C, Sowelam S, Sheldon TJ, et al. Reduction of right ventricular pacing in patients with sinus node dysfunction using an enhanced search AV algorithm. Pacing Clin Electrophysiol 2005; 28:521.
- Dennis MJ, Sparks PB. Pacemaker mediated tachycardia as a complication of the autointrinsic conduction search function. Pacing Clin Electrophysiol 2004; 27:824.
- Levine PA. First reported case where a PMT was repeatedly triggered by the AV. Pacing Clin Electrophysiol 2004; 27:1691.
- Sweeney MO, Ellenbogen KA, Casavant D, et al. Multicenter, prospective, randomized safety and efficacy study of a new atrial-based managed ventricular pacing mode (MVP) in dual chamber ICDs. J Cardiovasc Electrophysiol 2005; 16:811.
- Savouré A, Fröhlig G, Galley D, et al. A new dual-chamber pacing mode to minimize ventricular pacing. Pacing Clin Electrophysiol 2005; 28 Suppl 1:S43.
- Pascale P, Pruvot E, Graf D. Pacemaker syndrome during managed ventricular pacing mode: what is the mechanism? J Cardiovasc Electrophysiol 2009; 20:574.
- Barold SS, Stroobandt RX. Pacemaker-mediated tachycardia initiated by an atrioventricular search algorithm to minimize right ventricular pacing. J Electrocardiol 2012; 45:336.
- Himmrich E, Przibille O, Zellerhoff C, et al. Proarrhythmic effect of pacemaker stimulation in patients with implanted cardioverter-defibrillators. Circulation 2003; 108:192.
- Sweeney MO, Ruetz LL, Belk P, et al. Bradycardia pacing-induced short-long-short sequences at the onset of ventricular tachyarrhythmias: a possible mechanism of proarrhythmia? J Am Coll Cardiol 2007; 50:614.
- van Mechelen R, Schoonderwoerd R. Risk of managed ventricular pacing in a patient with heart block. Heart Rhythm 2006; 3:1384.
- Friedman PA, Jalal S, Kaufman S, et al. Effects of a rate smoothing algorithm for prevention of ventricular arrhythmias: results of the Ventricular Arrhythmia Suppression Trial (VAST). Heart Rhythm 2006; 3:573.
- Viskin S, Fish R, Zeltser D, et al. Arrhythmias in the congenital long QT syndrome: how often is torsade de pointes pause dependent? Heart 2000; 83:661.
- Predisposing factors
- INAPPROPRIATE MODE SWITCH
- PACEMAKER-INITIATED TACHYCARDIA
- PACEMAKER-MEDIATED TACHYCARDIA
- Tracking of a native atrial tachyarrhythmia
- Endless loop tachycardia
- Management of an endless loop tachycardia
- REPETITIVE NON-REENTRANT VENTRICULOATRIAL SYNCHRONOUS RHYTHMS (AV DESYNCHRONIZATION ARRHYTHMIA)
- FUNCTIONAL SINGLE CHAMBER ATRIAL PACING
- Long programmed AV delay
- AV/PV hysteresis
- AAI/DDD pacing
- VENTRICULAR ARRHYTHMIAS