Echocardiographic evaluation of prosthetic heart valves
- Elyse Foster, MD
Elyse Foster, MD
- Professor of Clinical Medicine
- University of California, San Francisco
- Section Editors
- Warren J Manning, MD
Warren J Manning, MD
- Section Editor — Noninvasive Cardiac Imaging and Stress Testing
- Professor of Medicine and Radiology
- Harvard Medical School
- William H Gaasch, MD
William H Gaasch, MD
- Section Editor — Valvular Disease
- Professor of Medicine
- University of Massachusetts Medical School
- Tufts University School of Medicine
- Senior Consultant in Cardiology
- Lahey Clinic
Echocardiography is the key noninvasive modality for evaluation of prosthetic valve structure and function . Transthoracic echocardiography (TTE) is the mainstay for monitoring prosthetic valves and can generally identify normal function as well as evidence of valve dysfunction (stenosis). Transesophageal echocardiography (TEE) is helpful particularly for assessment of valve structure and prosthetic valve regurgitation, especially involving mechanical mitral and tricuspid prostheses as well as assessment of endocarditis for all valves (figure 1).
This topic will review echocardiography of prosthetic heart valves. Management of patients with prosthetic heart valves and prosthetic valve complications are discussed separately. (See "Complications of prosthetic heart valves" and "Epidemiology, clinical manifestations, and diagnosis of prosthetic valve endocarditis" and "Antithrombotic therapy for prosthetic heart valves: Indications" and "Overview of the management of patients with prosthetic heart valves".)
Baseline transthoracic echocardiogram — We agree with the American Heart Association/American College of Cardiology (AHA/ACC) guideline recommendation to perform a transthoracic echocardiogram (TTE) six weeks to three months after valve implantation to evaluate valve hemodynamics and to establish a baseline for future comparison . The TTE should include Doppler measurements of transvalvular velocities as well as assessment of valvular and paravalvular regurgitation. Adequate Doppler velocity recordings can generally be obtained despite acoustic shadowing from valve prostheses.
Transvalvular gradients for normally functioning prosthetic valves are dependent upon valve type, location, and size (as compiled in the 2009 American Society of Echocardiography guidelines) as well as patient-specific factors . A higher than expected initial gradient is often due to a high output state (eg, due to anemia) or patient-prosthetic mismatch and is only rarely due to early dysfunction of the prosthesis. We suggest obtaining the postoperative baseline study after the patient’s hemoglobin has returned to baseline to avoid recording a gradient that is transiently higher than expected due to anemia.
Change in clinical status — The echocardiographer should be alert to the range of complications that can occur with prosthetic devices, including the following (see "Complications of prosthetic heart valves"):To continue reading this article, you must log in with your personal, hospital, or group practice subscription. For more information on subscription options, click below on the option that best describes you:
- Zoghbi WA, Chambers JB, Dumesnil JG, et al. Recommendations for evaluation of prosthetic valves with echocardiography and doppler ultrasound: a report From the American Society of Echocardiography's Guidelines and Standards Committee and the Task Force on Prosthetic Valves, developed in conjunction with the American College of Cardiology Cardiovascular Imaging Committee, Cardiac Imaging Committee of the American Heart Association, the European Association of Echocardiography, a registered branch of the European Society of Cardiology, the Japanese Society of Echocardiography and the Canadian Society of Echocardiography, endorsed by the American College of Cardiology Foundation, American Heart Association, European Association of Echocardiography, a registered branch of the European Society of Cardiology, the Japanese Society of Echocardiography, and Canadian Society of Echocardiography. J Am Soc Echocardiogr 2009; 22:975.
- Nishimura RA, Otto CM, Bonow RO, et al. 2014 AHA/ACC guideline for the management of patients with valvular heart disease: a report of the American College of Cardiology/American Heart Association Task Force on Practice Guidelines. J Am Coll Cardiol 2014; 63:e57.
- Chafizadeh ER, Zoghbi WA. Doppler echocardiographic assessment of the St. Jude Medical prosthetic valve in the aortic position using the continuity equation. Circulation 1991; 83:213.
- Cohen GI, Davison MB, Klein AL, et al. A comparison of flow convergence with other transthoracic echocardiographic indexes of prosthetic mitral regurgitation. J Am Soc Echocardiogr 1992; 5:620.
- Khandheria BK. Transesophageal echocardiography in the evaluation of prosthetic valves. Cardiol Clin 1993; 11:427.
- Faletra F, Constantin C, De Chiara F, et al. Incorrect echocardiographic diagnosis in patients with mechanical prosthetic valve dysfunction: correlation with surgical findings. Am J Med 2000; 108:531.
- Girard SE, Miller FA Jr, Orszulak TA, et al. Reoperation for prosthetic aortic valve obstruction in the era of echocardiography: trends in diagnostic testing and comparison with surgical findings. J Am Coll Cardiol 2001; 37:579.
- Ozkan M, Gündüz S, Yildiz M, Duran NE. Diagnosis of the prosthetic heart valve pannus formation with real-time three-dimensional transoesophageal echocardiography. Eur J Echocardiogr 2010; 11:E17.
- Deviri E, Sareli P, Wisenbaugh T, Cronje SL. Obstruction of mechanical heart valve prostheses: clinical aspects and surgical management. J Am Coll Cardiol 1991; 17:646.
- Lin SS, Tiong IY, Asher CR, et al. Prediction of thrombus-related mechanical prosthetic valve dysfunction using transesophageal echocardiography. Am J Cardiol 2000; 86:1097.
- Barbetseas J, Nagueh SF, Pitsavos C, et al. Differentiating thrombus from pannus formation in obstructed mechanical prosthetic valves: an evaluation of clinical, transthoracic and transesophageal echocardiographic parameters. J Am Coll Cardiol 1998; 32:1410.
- Baumgartner H, Czer L, DeRobertis M, et al. Normal regurgitation in mechanical valve prostheses: Impact on Doppler Studies. American College of Cardiology Learning Center Highlights 1992; 8:1.
- Baumgartner H, Khan S, DeRobertis M, et al. Color Doppler regurgitant characteristics of normal mechanical mitral valve prostheses in vitro. Circulation 1992; 85:323.
- Mohr-Kahaly S, Kupferwasser I, Erbel R, et al. Regurgitant flow in apparently normal valve prostheses: improved detection and semiquantitative analysis by transesophageal two-dimensional color-coded Doppler echocardiography. J Am Soc Echocardiogr 1990; 3:187.
- Karalis DG, Chandrasekaran K, Ross JJ Jr, et al. Single-plane transesophageal echocardiography for assessing function of mechanical or bioprosthetic valves in the aortic valve position. Am J Cardiol 1992; 69:1310.
- Lange HW, Olson JD, Pedersen WR, et al. Transesophageal color Doppler echocardiography of the normal St. Jude Medical mitral valve prosthesis. Am Heart J 1991; 122:489.
- Flachskampf FA, O'Shea JP, Griffin BP, et al. Patterns of normal transvalvular regurgitation in mechanical valve prostheses. J Am Coll Cardiol 1991; 18:1493.
- Hixson CS, Smith MD, Mattson MD, et al. Comparison of transesophageal color flow Doppler imaging of normal mitral regurgitant jets in St. Jude Medical and Medtronic Hall cardiac prostheses. J Am Soc Echocardiogr 1992; 5:57.
- Chen YT, Kan MN, Chen JS, et al. Detection of prosthetic mitral valve leak: a comparative study using transesophageal echocardiography, transthoracic echocardiography, and auscultation. J Clin Ultrasound 1990; 18:557.
- Fernandes V, Olmos L, Nagueh SF, et al. Peak early diastolic velocity rather than pressure half-time is the best index of mechanical prosthetic mitral valve function. Am J Cardiol 2002; 89:704.
- Dilip KA, Vachaspathy P, Clarke B, et al. Haemolysis following mitral valve repair. J Cardiovasc Surg (Torino) 1992; 33:568.
- Smith RE, Berg D. Occult paravalvular leak in a clinically normal St. Jude's mitral valve presenting with life-threatening microangiopathic hemolytic anemia. J Cardiovasc Surg (Torino) 1991; 32:56.
- Groundstroem K, Rittoo D, Hoffman P, et al. Additional value of biplane transoesophageal imaging in assessment of mitral valve prostheses. Br Heart J 1993; 70:259.
- Akamatsu S, Ueda N, Terazawa E, et al. Mitral prosthetic dehiscence with laminar regurgitant flow signals assessed by transesophageal echocardiography. Chest 1993; 104:1911.
- Kronzon I, Sugeng L, Perk G, et al. Real-time 3-dimensional transesophageal echocardiography in the evaluation of post-operative mitral annuloplasty ring and prosthetic valve dehiscence. J Am Coll Cardiol 2009; 53:1543.
- Shiran A, Weissman NJ, Merdler A, et al. Transesophageal echocardiographic findings in patients with nonobstructed prosthetic valves and suspected cardiac source of embolism. Am J Cardiol 2001; 88:1441.
- Pibarot P, Dumesnil JG. Hemodynamic and clinical impact of prosthesis-patient mismatch in the aortic valve position and its prevention. J Am Coll Cardiol 2000; 36:1131.
- Blauwet LA, Danielson GK, Burkhart HM, et al. Comprehensive echocardiographic assessment of the hemodynamic parameters of 285 tricuspid valve bioprostheses early after implantation. J Am Soc Echocardiogr 2010; 23:1045.
- Blauwet LA, Burkhart HM, Dearani JA, et al. Comprehensive echocardiographic assessment of mechanical tricuspid valve prostheses based on early post-implantation echocardiographic studies. J Am Soc Echocardiogr 2011; 24:414.
- MONITORING GUIDELINES
- Baseline transthoracic echocardiogram
- Change in clinical status
- Surveillance of old bioprosthetic valves
- KEY COMPONENTS OF THE ECHOCARDIOGRAM
- Role of TTE and TEE
- FEATURES OF VALVE DYSFUNCTION
- Prosthetic valve obstruction
- - Distinction between thrombus and pannus
- Prosthetic valve regurgitation
- - Physiologic regurgitation
- - Pathologic regurgitation
- Paravalvular regurgitation
- Prosthetic valve dehiscence
- SPECIFIC PROSTHETIC VALVE DISORDERS
- Prosthetic aortic stenosis
- Prosthetic aortic regurgitation
- Prosthetic mitral stenosis
- Prosthetic mitral regurgitation
- Prosthetic pulmonary valves dysfunction
- Prosthetic tricuspid valve dysfunction
- SUMMARY AND RECOMMENDATIONS