Pulmonary artery catheterization: Interpretation of hemodynamic values and waveforms in adults
- Frank E Silvestry, MD
Frank E Silvestry, MD
- Associate Professor of Medicine
- Hospital of the University of Pennsylvania
The pulmonary artery catheter (PAC; Swan-Ganz or right heart catheter) can be used for a variety of clinical purposes. Interpreting hemodynamic data from PACs is important for the diagnosis and management of a range of conditions including shock and pulmonary artery hypertension (table 1).
The interpretation of hemodynamic values and pressure tracings derived from the PAC is described in this topic. The insertion technique, indications, contraindications, and complications of PACs are discussed separately. (See "Pulmonary artery catheters: Insertion technique in adults" and "Pulmonary artery catheterization: Indications, contraindications, and complications in adults".)
Direct measurements of the following can be obtained from an accurately placed pulmonary artery catheter (PAC):
●Central venous pressure (CVP)
●Right-sided intracardiac pressures (right atrium, right ventricle)
Subscribers log in hereTo continue reading this article, you must log in with your personal, hospital, or group practice subscription. For more information or to purchase a personal subscription, click below on the option that best describes you:Literature review current through: May 2017. | This topic last updated: Dec 11, 2015.References
- Kovacs G, Avian A, Olschewski A, Olschewski H. Zero reference level for right heart catheterisation. Eur Respir J 2013; 42:1586.
- Kleinman B, Powell S, Kumar P, Gardner RM. The fast flush test measures the dynamic response of the entire blood pressure monitoring system. Anesthesiology 1992; 77:1215.
- Swan HJ, Ganz W, Forrester J, et al. Catheterization of the heart in man with use of a flow-directed balloon-tipped catheter. N Engl J Med 1970; 283:447.
- Weed HG. Pulmonary "capillary" wedge pressure not the pressure in the pulmonary capillaries. Chest 1991; 100:1138.
- O'Quin R, Marini JJ. Pulmonary artery occlusion pressure: clinical physiology, measurement, and interpretation. Am Rev Respir Dis 1983; 128:319.
- Putterman, C. The Swan-Ganz catheter: A decade of hemodynamic monitoring. J Crit Care 1989; 4:127.
- Sharkey SW. Beyond the wedge: clinical physiology and the Swan-Ganz catheter. Am J Med 1987; 83:111.
- Raper R, Sibbald WJ. Misled by the wedge? The Swan-Ganz catheter and left ventricular preload. Chest 1986; 89:427.
- Nemens EJ, Woods SL. Normal fluctuations in pulmonary artery and pulmonary capillary wedge pressures in acutely ill patients. Heart Lung 1982; 11:393.
- Snyder RW 2nd, Glamann DB, Lange RA, et al. Predictive value of prominent pulmonary arterial wedge V waves in assessing the presence and severity of mitral regurgitation. Am J Cardiol 1994; 73:568.
- Pinsky MR. Pulmonary artery occlusion pressure. Intensive Care Med 2003; 29:19.
- Pinsky MR. Clinical significance of pulmonary artery occlusion pressure. Intensive Care Med 2003; 29:175.
- WEST JB, DOLLERY CT, NAIMARK A. DISTRIBUTION OF BLOOD FLOW IN ISOLATED LUNG; RELATION TO VASCULAR AND ALVEOLAR PRESSURES. J Appl Physiol 1964; 19:713.
- Teboul, JL, Besbes, M, Andrivet, P, et al. A bedside index assessing the reliability of pulmonary artery occlusion pressure measurements during mechanical ventilation with positive end-expiratory pressure. J Crit Care 1992; 7:22.
- Yelderman ML, Ramsay MA, Quinn MD, et al. Continuous thermodilution cardiac output measurement in intensive care unit patients. J Cardiothorac Vasc Anesth 1992; 6:270.
- Armaganidis A, Dhainaut JF, Billard JL, et al. Accuracy assessment for three fiberoptic pulmonary artery catheters for SvO2 monitoring. Intensive Care Med 1994; 20:484.
- London MJ, Moritz TE, Henderson WG, et al. Standard versus fiberoptic pulmonary artery catheterization for cardiac surgery in the Department of Veterans Affairs: a prospective, observational, multicenter analysis. Anesthesiology 2002; 96:860.
- Gore, JM, Alpert, JS, Benotti, JR, et al. Handbook of Hemodynamic Monitoring, 1st, Little Brown, Boston 1985.
- Cardiac Catheterization and Angiography, 4th, Grossman, W (Eds), Lea & Febiger, Philadelphia 1996.
- Flamm MD, Cohn KE, Hancock EW. Measurement of systemic cardiac output at rest and exercise in patients with atrial septal defect. Am J Cardiol 1969; 23:258.
- PHYSIOLOGIC MEASUREMENTS
- ENSURING ACCURATE MEASUREMENTS
- Zeroing and referencing
- Correct placement
- The "fast flush" test
- CATHETER WAVEFORMS AND PRESSURES
- Right atrium (RA)
- - Elevated RA pressure
- - Abnormal RA waveforms
- Right ventricle
- Pulmonary artery
- Pulmonary artery occlusion pressure (PAOP)
- - Abnormal PAOP
- - Abnormal PAOP waveforms
- - Sources of error
- Discordant PAOP and LV end-diastolic pressure
- Lung zone misplacement
- Respiratory effects
- Positive end-expiratory pressure
- CALCULATION OF CARDIAC OUTPUT
- Indicator thermodilution method
- Fick method
- DETECTION OF LEFT-TO-RIGHT SHUNTS
- SYSTEMIC AND PULMONARY VASCULAR RESISTANCE
- SUMMARY AND RECOMMENDATIONS