Portable oxygen delivery and oxygen conserving devices
- Brian L Tiep, MD
Brian L Tiep, MD
- Director of Pulmonary Rehabilitation, City of Hope National Medical Center
- Associate Professor of Medicine, Western University of Health Sciences
- Rick Carter, PhD, MBA
Rick Carter, PhD, MBA
- Lamar University, Beaumont, TX
Patients with chronic lung disease and other conditions who require long-term oxygen therapy are often mobile outside of the home and need access to a portable supply of oxygen to maintain a normal lifestyle. For patients with advanced lung disease who require long-term supplemental oxygen, exertional breathlessness, hypoxemia, and lack of energy are compounded by the increased demand of carrying or dragging portable oxygen equipment when they are away from home.
Portable oxygen sources can be heavy, cumbersome, and limited in the duration of oxygen supply, so oxygen conserving devices have been introduced as a means of making oxygen therapy more efficient, more portable, and less intrusive [1,2].
This review will compare traditional, continuous flow oxygen delivery by nasal cannula with a variety of oxygen conserving devices. The indications for long-term supplemental oxygen, the use of oxygen in hypercapnic patients, and issues regarding oxygen therapy during air travel are discussed separately. (See "Long-term supplemental oxygen therapy" and "The evaluation, diagnosis, and treatment of the adult patient with acute hypercapnic respiratory failure" and "Traveling with oxygen aboard commercial air carriers".)
CONTINUOUS FLOW NASAL CANNULA
Continuous flow oxygen delivery through nasal cannula is the usual prescription for long-term oxygen delivery in hypoxemic patients and, thus, is the standard against which all oxygen-conserving techniques should be compared . An important consideration for patients who are hypoxemic at rest is that their oxygen demands increase, sometimes abruptly, with exertion such as during activities of daily living, walking, and climbing stairs. As a result, higher oxygen flows are often necessary during exertion. Engineering technologies are making portable high-flow delivery possible for those individuals requiring high-flow supplemental oxygen. (See "Long-term supplemental oxygen therapy", section on 'Prescribing oxygen'.)
While the combination of continuous flow and the nasal cannula interface is widely used for long-term supplemental oxygen, the system is inefficient, as only a small percentage of the oxygen delivered to the nose actually reaches the alveoli. Oxygen flowing through the standard nasal cannula is nearly 100 percent pure. However, the patient actually receives a blend of pure oxygen and room air, as atmospheric air (20.9 percent oxygen) is entrained with the oxygen . The resultant inspiratory oxygen concentrations of standard nasal cannula flow settings are approximated in the figure (figure 1).
- McCoy R. Oxygen-conserving techniques and devices. Respir Care 2000; 45:95.
- Keller RR. Long-term oxygen therapy: advances and perspectives in technical devices. Monaldi Arch Chest Dis 1999; 54:75.
- Rees PJ, Dudley F. Provision of oxygen at home. BMJ 1998; 317:935.
- Tiep BL. Continuous flow oxygen therapy and basis for improving the efficiency of oxygen delivery. In: Portable Oxygen Therapy: Including Oxygen Conserving Methodology, Tiep BL (Ed), Futura Publishing, Mt Kisco, New York 1991. p.205.
- Mithoefer JC, Keighley JF, Karetzky MS. Response of the arterial Po2 to oxygen administration in chronic pulmonary disease. Interpretation of findings in a study of 46 patients and 14 normal subjects. Ann Intern Med 1971; 74:328.
- Soffer M, Tashkin DP, Shapiro BJ, et al. Conservation of oxygen supply using a reservoir nasal cannula in hypoxemic patients at rest and during exercise. Chest 1985; 88:663.
- Gloeckl R, Heinzelmann I, Matthaei M, et al. Benefits of an oxygen reservoir cannula versus a conventional nasal cannula during exercise in hypoxemic COPD patients: a crossover trial. Respiration 2014; 88:399.
- Gonzales, S, Huntington, D, Remo, R, Light, R. Efficacy of the Oxymizer Pendant in reducing oxygen requirements of hypoxemic patients. Respir Care 1986; 31:681.
- Tiep BL, Nicotra B, Carter R, et al. Evaluation of a low-flow oxygen-conserving nasal cannula. Am Rev Respir Dis 1984; 130:500.
- Pesce, L, Lauro, S, Testi, R, et al. Effetti della somministrazione de ossigeno attraverso un nuovo dispositivo economizzatore in 20 pazienti affetti da insufficcienza respiratoria. Min Pneum 1987; 25:143.
- Carter R, Williams JS, Berry J, et al. Evaluation of the pendant oxygen-conserving nasal cannula during exercise. Chest 1986; 89:806.
- Hagarty EM, Skorodin MS, Stiers WM, et al. Performance of a reservoir nasal cannula (Oxymizer) during sleep in hypoxemic patients with COPD. Chest 1993; 103:1129.
- Collard P, Wautelet F, Delwiche JP, et al. Improvement of oxygen delivery in severe hypoxaemia by a reservoir cannula. Eur Respir J 1989; 2:778.
- Cuvelier A, Muir JF, Czernichow P, et al. Nocturnal efficiency and tolerance of a demand oxygen delivery system in COPD patients with nocturnal hypoxemia. Chest 1999; 116:22.
- Heimlich HJ. Respiratory rehabilitation with transtracheal oxygen system. Ann Otol Rhinol Laryngol 1982; 91:643.
- Christopher KL, Spofford BT, Petrun MD, et al. A program for transtracheal oxygen delivery. Assessment of safety and efficacy. Ann Intern Med 1987; 107:802.
- Christopher KL, Spofford BT, Brannin PK, Petty TL. Transtracheal oxygen therapy for refractory hypoxemia. JAMA 1986; 256:494.
- Hoffman LA, Wesmiller SW, Sciurba FC, et al. Nasal cannula and transtracheal oxygen delivery. A comparison of patient response after 6 months of each technique. Am Rev Respir Dis 1992; 145:827.
- Benditt J, Pollock M, Roa J, Celli B. Transtracheal delivery of gas decreases the oxygen cost of breathing. Am Rev Respir Dis 1993; 147:1207.
- Tiep BL, Christopher KL, Spofford BT, et al. Pulsed nasal and transtracheal oxygen delivery. Chest 1990; 97:364.
- Yaeger ES, Goodman S, Hoddes E, Christopher KL. Oxygen therapy using pulse and continuous flow with a transtracheal catheter and a nasal cannula. Chest 1994; 106:854.
- Tiep BL, Nicotra MB, Carter R, et al. Low-concentration oxygen therapy via a demand oxygen delivery system. Chest 1985; 87:636.
- Porszasz J, Cao R, Morishige R, et al. Physiologic effects of an ambulatory ventilation system in chronic obstructive pulmonary disease. Am J Respir Crit Care Med 2013; 188:334.
- Garrod R, Bestall JC, Paul E, Wedzicha JA. Evaluation of pulsed dose oxygen delivery during exercise in patients with severe chronic obstructive pulmonary disease. Thorax 1999; 54:242.
- Carter R, Tashkin D, Djahed B, et al. Demand oxygen delivery for patients with restrictive lung disease. Chest 1989; 96:1307.
- Franco, MA, Conner, SA, Gougenheim, C. Pulse dose oxygen delivery during exercise. Respir Care 1985; 30:888.
- Bower JS, Brook CJ, Zimmer K, Davis D. Performance of a demand oxygen saver system during rest, exercise, and sleep in hypoxemic patients. Chest 1988; 94:77.
- Fuhrman C, Chouaid C, Herigault R, et al. Comparison of four demand oxygen delivery systems at rest and during exercise for chronic obstructive pulmonary disease. Respir Med 2004; 98:938.
- Tiep BL, Barnett J, Schiffman G, et al. Maintaining oxygenation via demand oxygen delivery during rest and exercise. Respir Care 2002; 47:887.
- Palwai A, Skowronski M, Coreno A, et al. Critical comparisons of the clinical performance of oxygen-conserving devices. Am J Respir Crit Care Med 2010; 181:1061.
- Martí S, Pajares V, Morante F, et al. Are oxygen-conserving devices effective for correcting exercise hypoxemia? Respir Care 2013; 58:1606.
- Tiep, BL, Murray, R, Barnett, M, Carter, R. Auto-adjusting demand oxygen delivery system that minimizes SaO2 swings between rest and exertion. Chest 2004 (Abstract); 126:763S.
- Rice KL, Schmidt MF, Buan JS, et al. AccuO2 oximetry-driven oxygen-conserving device versus fixed-dose oxygen devices in stable COPD patients. Respir Care 2011; 56:1901.
- CONTINUOUS FLOW NASAL CANNULA
- METHODS OF OXYGEN CONSERVATION
- Reservoir cannulas
- Transtracheal catheters
- Demand oxygen pulse devices
- - How demand oxygen pulse devices work
- - Types of devices
- - Variability in oxygenation
- - Future innovations
- PRESCRIBING OXYGEN CONSERVING DEVICES
- PORTABLE OXYGEN SYSTEMS
- Lightweight compressed gas cylinders
- Liquid oxygen systems
- Portable oxygen concentrators
- PRESCRIBING PORTABLE OXYGEN SYSTEMS
- SUMMARY AND RECOMMENDATIONS