Plasmapheresis with hemodialysis equipment
- Andre A Kaplan, MD
Andre A Kaplan, MD
- Professor of Medicine
- University of Connecticut Health Center
Traditionally, therapeutic plasma exchange (TPE, plasmapheresis) was performed with centrifugation devices used in blood blanking procedures. (See "Therapeutic apheresis (plasma exchange or cytapheresis): Indications and technology".)
These devices permit selective cell removal (cytapheresis) but have been associated with posttreatment thrombocytopenia  and require a dedicated centrifuge device. An increasingly popular and often more efficient alternative is to perform plasma exchange by membrane plasma separation (MPS) [2-5].
MPS utilizes a highly permeable membrane and a standard dialysis machine used in its ultrafiltration mode, dialysis-bypass mode (similar to the technique in hemoperfusion). TPE is also possible using NxStage and Prismaflex continuous renal replacement therapy (CRRT) machines. The NxStage machine employs a cartridge that uses the Plasmaflo OP-05H as the filtration membrane. (See 'Technical considerations' below and "Short daily home hemodialysis: The low dialysate volume approach" and "Continuous venovenous hemodialysis: Technical considerations".)
For many years, the most commonly used membrane in the United States was the Plasmaflo AP-05H. This filter has been replaced with a slightly more porous filter, the OP-05W . This filter can be used with most dialysis machines, but the Cobe 3 and Hospal machines are incompatible with the filter's tubing.
The published American experience with MPS is very limited even though thousands of treatments are performed in the United States each year [7-9]. In some countries, such as Germany and Japan, 90 percent of plasmapheresis treatments are performed by membrane plasma separation [5,10].To continue reading this article, you must log in with your personal, hospital, or group practice subscription. For more information on subscription options, click below on the option that best describes you:
- Sowada K, Malchesky PS, Nose Y. Available removal systems: State of the art. In: Therapeutic Hemapheresis in the 1990s, Nydegger UE (Ed), Karger, Basel 1990. Vol 57, p.51.
- Gurland HJ, Lysaght MJ, Samtleben W, Schmidt B. A comparison of centrifugal and membrane-based apheresis formats. Int J Artif Organs 1984; 7:35.
- Gurland HJ, Lysaght MJ, Samtleben W, Schmidt B. Comparative evaluation of filters used in membrane plasmapheresis. Nephron 1984; 36:173.
- Kaplan AA. Therapeutic plasma exchange: core curriculum 2008. Am J Kidney Dis 2008; 52:1180.
- Kaplan AA. Why nephrologists should do therapeutic plasma exchange. Dial Transplant 2009; 38:65.
- Nagano T, Nakazono K, Iwamoto H, et al. Comparative study of polyethylene plasma separator sterilized by ethylene oxide and gamma rays. Jpn J Apheresis 1997; 16:297.
- Gerhardt RE, Ntoso KA, Koethe JD, et al. Acute plasma separation with hemodialysis equipment. J Am Soc Nephrol 1992; 2:1455.
- Howard RL, Tatum K, Chan L, Shapiro JI. Performance of plasmapheresis by nephrologists: A survey. Dial Transplant 1990; 19:484.
- Grandham S, Yousaf F, Win Y, et al. Therapeutic plasmapheresis in primary presentation of diabetes mellitus with diabetic ketoacidosis, hypertriglyceridemia and acute pancreatitis. World J Nephrol Urol 2014; 3:162.
- Malchesky PS, Koo AP, Roberson GA, et al. Apheresis technologies and clinical applications: the 2005 International Apheresis Registry. Ther Apher Dial 2007; 11:341.
- Price CA. Therapeutic plasma exchange in a dialysis unit. ANNA J 1987; 14:103.
- Mokrzycki MH, Kaplan AA. Therapeutic plasma exchange: complications and management. Am J Kidney Dis 1994; 23:817.
- Price CA, McCarley PB. Technical considerations of therapeutic plasma exchange as a nephrology nursing procedure. ANNA J 1993; 20:41.
- Price CA, McCarley PB. Physical assessment for patients receiving therapeutic plasma exchange. ANNA J 1994; 21:149.
- Ilamathi E, Kirsch M, Moore B, Finger M. Citrate anticoagulation during plasmapheresis using standard hemodialysis equipment. Semin Dial 1993; 6:268.
- Frascà GM, Buscaroli A, Borgnino LC, Vangelista A. Optimization of heparin anticoagulation during membrane plasma separation. Int J Artif Organs 1988; 11:313.
- Pearl RG, Rosenthal MH. Metabolic alkalosis due to plasmapheresis. Am J Med 1985; 79:391.
- Kaplan AA, Halley SE. Plasma exchange with a rotating filter. Kidney Int 1990; 38:160.