Endotracheal tube introducers (gum elastic bougie) for emergency intubation
- Aaron E Bair, MD, MSc, FAAEM, FACEP
Aaron E Bair, MD, MSc, FAAEM, FACEP
- Professor of Emergency Medicine
- University of California, Davis
- Erik G Laurin, MD, FAAEM
Erik G Laurin, MD, FAAEM
- Professor of Emergency Medicine
- University of California, Davis, Medical Center
The endotracheal tube introducer (ETI) is an effective and inexpensive adjunct to difficult airway management that is easy to use. We recommend that an ETI be readily available in every emergency department.
This topic will review the types of ETIs, indications and contraindications for their use, proper technique for using the devices during emergency tracheal intubation, and evidence of their effectiveness. Other airway devices and aspects of emergency airway management are discussed separately. (See "Devices for difficult emergency airway management in adults" and "Advanced emergency airway management in adults" and "Rapid sequence intubation for adults outside the operating room" and "Emergency airway management in children: Unique pediatric considerations".)
TERMINOLOGY AND EQUIPMENT
Several terms are used to describe the classic endotracheal tube introducer (ETI). Although the phrase "gum elastic bougie" is common, we find it confusing since the ETI is neither gum nor elastic and is not used as a bougie (ie, dilator). In this topic, we will refer to the device as an endotracheal or tracheal tube introducer.
The ETI consists of a 60 cm stylet with the distal tip bent at a 30 degree angle. The bend allows the intubator to direct the tip anteriorly under the epiglottis and through the vocal cords, which may not be visible.
Dimensions and use are comparable among the three main types of ETI available:
- Kidd JF, Dyson A, Latto IP. Successful difficult intubation. Use of the gum elastic bougie. Anaesthesia 1988; 43:437.
- Bair AE, Laurin EG, Schmitt BJ. An assessment of a tracheal tube introducer as an endotracheal tube placement confirmation device. Am J Emerg Med 2005; 23:754.
- Marson BA, Anderson E, Wilkes AR, Hodzovic I. Bougie-related airway trauma: dangers of the hold-up sign. Anaesthesia 2014; 69:219.
- Gataure PS, Vaughan RS, Latto IP. Simulated difficult intubation. Comparison of the gum elastic bougie and the stylet. Anaesthesia 1996; 51:935.
- Detave M, Shiniara M, Leborgne JM. [Use of Eschmann's gum elastic bougie in difficult orotracheal intubation, an audit over eight years of clinical practice]. Ann Fr Anesth Reanim 2008; 27:154.
- Martin LD, Mhyre JM, Shanks AM, et al. 3,423 emergency tracheal intubations at a university hospital: airway outcomes and complications. Anesthesiology 2011; 114:42.
- Le DH, Reed DB, Weinstein G, et al. Paramedic use of endotracheal tube introducers for the difficult airway. Prehosp Emerg Care 2001; 5:155.
- Phelan MP, Moscati R, D'Aprix T, Miller G. Paramedic use of the endotracheal tube introducer in a difficult airway model. Prehosp Emerg Care 2003; 7:244.
- Jabre P, Combes X, Leroux B, et al. Use of gum elastic bougie for prehospital difficult intubation. Am J Emerg Med 2005; 23:552.
- Kadry M, Popat M. Pharyngeal wall perforation--an unusual complication of blind intubation with a gum elastic bougie. Anaesthesia 1999; 54:404.
- Kadry T, Harvey M, Wallace M, Imrie J. Frova intubating catheter position can be determined with aspirating oesophageal detection device. Emerg Med Australas 2007; 19:203.
- Takenaka I, Aoyama K, Iwagaki T, et al. Approach combining the airway scope and the bougie for minimizing movement of the cervical spine during endotracheal intubation. Anesthesiology 2009; 110:1335.
- Inoue H, Saito T, Kamishima K, et al. [Successful nasal intubation using airway scope with gum elastic bougie in a case of difficult airway]. Masui 2008; 57:457.
- Rich JM. Successful blind digital intubation with a bougie introducer in a patient with an unexpected difficult airway. Proc (Bayl Univ Med Cent) 2008; 21:397.