Echocardiography essentials: Physics and instrumentation
- Warren J Manning, MD
Warren J Manning, MD
- Section Editor — Noninvasive Cardiac Imaging and Stress Testing
- Professor of Medicine and Radiology
- Harvard Medical School
Echocardiography is the primary noninvasive imaging modality for quantitative and qualitative evaluation of cardiac anatomy, physiology, and function [1,2]. Two-dimensional (2D) echocardiography provides tomographic or "thin-slice" imaging. Comprehensive echocardiographic examination typically involves imaging the heart from multiple "viewing" orientations as well as with multiple imaging techniques (ie, 2D, Doppler). An understanding of the fundamental principles of cardiac ultrasound is necessary for the proper acquisition and interpretation of echocardiographic data.
The basic physics principles that apply to echocardiography and its instrumentation are reviewed here. The normal anatomic views and protocols for echocardiography, as well as its clinical uses, are presented separately. (See "Transthoracic echocardiography: Normal cardiac anatomy and tomographic views".)
Sound waves are mechanical vibrations that can be described in terms of frequency or Hertz (Hz), ie, the number of repetitions or cycles per second. Other characteristics include wavelength, the distance between excitations, measured in mm; and the amplitude of excitation, measured in decibels (dB). A 6-dB change results in a doubling (or halving) of the signal amplitude.
Medical ultrasound imaging typically uses sound waves at frequencies of 1,000,000 to 20,000,000 Hz (1.0 to 20 MHz). In contrast, the human auditory spectrum comprises frequencies between 20 and 20,000 Hz.
Frequency and wavelength are mathematically related to the velocity of the ultrasound beam within the tissue (approximately 1,540,000 mm/sec for human tissue) as indicated by the following equations:
- Pearlman JD, Triulzi MO, King ME, et al. Limits of normal left ventricular dimensions in growth and development: analysis of dimensions and variance in the two-dimensional echocardiograms of 268 normal healthy subjects. J Am Coll Cardiol 1988; 12:1432.
- Otto CM. Textbook of Clinical Echocardiography, 4th edition, Saunders Elsevier, 2009.
- Caidahl K, Kazzam E, Lidberg J, et al. New concept in echocardiography: harmonic imaging of tissue without use of contrast agent. Lancet 1998; 352:1264.
- Spencer KT, Bednarz J, Rafter PG, et al. Use of harmonic imaging without echocardiographic contrast to improve two-dimensional image quality. Am J Cardiol 1998; 82:794.
- Main ML, Asher CR, Rubin DN, et al. Comparison of tissue harmonic imaging with contrast (sonicated albumin echocardiography and Doppler myocardial imaging for enhancing endocardial border resolution. Am J Cardiol 1999; 83:218.
- Franke A, Hoffmann R, Kühl HP, et al. Non-contrast second harmonic imaging improves interobserver agreement and accuracy of dobutamine stress echocardiography in patients with impaired image quality. Heart 2000; 83:133.
- Prinz C, Voigt JU. Diagnostic accuracy of a hand-held ultrasound scanner in routine patients referred for echocardiography. J Am Soc Echocardiogr 2011; 24:111.
- Lipczyńska M, Szymański P, Klisiewicz A, Hoffman P. Hand-carried echocardiography in heart failure and heart failure risk population: a community based prospective study. J Am Soc Echocardiogr 2011; 24:125.
- Spencer KT, Anderson AS, Bhargava A, et al. Physician-performed point-of-care echocardiography using a laptop platform compared with physical examination in the cardiovascular patient. J Am Coll Cardiol 2001; 37:2013.
- Cardim N, Fernandez Golfin C, Ferreira D, et al. Usefulness of a new miniaturized echocardiographic system in outpatient cardiology consultations as an extension of physical examination. J Am Soc Echocardiogr 2011; 24:117.
- Weiner RB, Wang F, Hutter AM Jr, et al. The feasibility, diagnostic yield, and learning curve of portable echocardiography for out-of-hospital cardiovascular disease screening. J Am Soc Echocardiogr 2012; 25:568.
- ULTRASOUND WAVES
- INTERACTION OF ULTRASOUND WAVES WITH TISSUES
- ULTRASOUND TRANSDUCERS
- Second harmonic imaging
- IMAGING MODALITIES
- Two-dimensional (2D) imaging
- Miniaturized echocardiography
- BIOEFFECTS AND SAFETY
- Thermal effects
- SUMMARY AND RECOMMENDATIONS